How do I get instrument training for XRD?
Please follow the instructions on this wiki-page.
How do I get access to the data and the XRD software?
After you have participated in the instrument training session, you get an email with some information and passwords etc (Subject: Getting started with XRD analysis). You can also follow the descriptions here.
How do I use the database?
There are currently 2 options:
1) PDF (Powder Diffraction File): On the lab PCs we have licenses for the ICDD-4+PDF-database. A quick demo of this database can be viewed here: https://www.youtube.com/user/TheICDD.
2) COD (Crystallography Open Database): http://www.crystallography.net/cod/. This database can be installed on your own computer, and you can implement it in EVA for Search/Match. "Tools" → "Settings" → "Database", and select "Crystallography Open Database".
You can use the databases directly (e.g. before measuring) to seach for patterns, however, they are also implemented in the DIFFRAC.EVA-software. First you have to download the "cod-for-diffraceva_vxx.exe" (see the "Getting started with XRD analysis-email descriped above for details), and install the .exe-file on your computer. Then open EVA, and click "Tool" → "Search / Match". For more details, please see the "EVA-How to" booklet in the lab.
Routine powder diffractometer (DaVinci1)
I measure some small peaks that I can't identify. What can it be?
- If the small peaks are at slightly smaller 2-theta-angle than the main peaks (and measured at the routine diffractometer (DaVinci1)), it is perhaps W-La or Cu-Kbeta contamination from the X-ray source. For explanation, please log on to the odin server, and select "/useful_documents/d8_davinci_1/Explanation of the peaks due to different wavelengths.pdf" .
- How to match the contamination-peaks in EVA: please log on to the odin server, and select "/useful_documents/d8_davinci_1/Identifying peaks due to different wavelength.pdf" .
Have my samples been measured? I came to the lab to check, however, I didn’t find the new files in my folder.
Check if the files have been saved to the default directory: C:\ProgramData\Bruker AXS\Results
9-position sample holder powder diffractometer (D8-Focus)
I get an "error" when I try to start my measurements: "some drives need to be initialized. Check for crashing possibilities". What should I do?
It is not really an error, just a warning. Simply click "OK", and watch the diffractometer arms/sample changer as the motors move.
What is the maximum size a sample can be for standard deep sample holders?
The sample has to fit within the sample holder that is 5 mm deep, and 40 mm diameter, as shown is these photos.
|Top view||Side view|